View and Download ABB Advance Optima Uras 14 service manual online. Advance Optima Uras 14 Measuring Instruments pdf manual download. Manuals and User Guides for ABB Advance Optima Uras We have 1 ABB Advance Optima Uras 14 manual available for free PDF download: Service Manual. Spare Parts for ABB & Siemens Gas Analyzer & Sample Handling Systems is available. Home» Products» ABB Gas Analyzer Spare Parts» Module Uras
|Country:||Papua New Guinea|
|Published (Last):||23 June 2015|
|PDF File Size:||1.39 Mb|
|ePub File Size:||13.97 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 95 Chapter 6: Page Pressure detector initial calibration, continued Performing the initial The pressure detector is only calibrated at the start point. Comments to this Manuals Your Name.
Emitter insert electrical passages Where? Detector configuration Detector Detector type: Please enter up to 7 characters for the postcode.
Pump General As an option, an internal pump can be installed to supply the sample gas. After calibration messages 1, 2, 4, 8 or 16 may appear.
Pay with PayPal and you’re fully protected. Abb Error Temperature error Indication Desired detector temperature not reached. Calibration Cell Calibration cell Location in module The calibration cell is a option to be mounted in the calibration unit.
The main frame always has two apertures installed and these are moved back and Structure forth horizontally by a screw-type drive mechanism. Please enter a valid postcode. Enter “0” no flow Enter the desired span point value: Analyzer module versions Overview This chapter describes the individual module versions.
ABB Advance Optima Uras 14 Manuals
Page IR detectors, continued Measurement range For each component, 2 measurement ranges can be configured. Page 27 Module versions plastic tubing connectionscontinued Figure Analysatormodul Uras 14 Pneumatikmodul Module versions: Physical measurement value; regulator value Explanation: Select a valid country.
The module is generally equipped with hardware for the thermostat version. Ruas Circuit Board Thermostat circuit board Location in module The circuit board is attached to the main frame.
Don’t show me this message again. Page 6 Contents, continued Page Chapter 5: IR detectors, continued Supplementary help Measurement ranges from – to Start and end of the reference measurement range Autorange possible Click the “Measurement range” button to enable autoranging for the measurement range Correction functions Correction value selection at the measurement range level See below for explanation.
Page of Go.
ABB ADVANCE OPTIMA URAS 14 SERVICE MANUAL Pdf Download.
If only one emitter insert is used, it can be placed in the first or second beam path depending on the requirements of the measurement application. Design For use in Zone 1 and Zone 2 combustible measurement gas the analyzer module is installed in a pressure-tight cylinder. Heater Accessories Multimeter Heating resistor resistance Electrical value: Infrared detector, continued Figure Infrared detector 1 Amplifier matching jumpers 2 Preamplifier 3 IR module circuit board plug connection 4 Membrane capacitor 5 Front and rear chambers6 Positioning pins and openings 7 Fill ports sealed gas-tight and soldered Figure Preamplifier BR 1 1x amplification The pressure sensor measures pressure in the sample cell.
The curve is calculated at 17 predetermined set points. Find out more about your rights as a buyer – opens in a new window or tab and exceptions – opens in a new window or tab. Please allow additional time if international delivery is subject to customs processing. Page IR detectors, continued Detector The data in the following tables are guidelines. Questions and answers about this item.
Change abg Note The modulator can be changed without removing the analyzer module. International postage and import charges paid to Pitney Bowes Inc. Module versions Udas Module versions principles Module versions plastic tubing connections Bab versions stainless steel tubing Ex analyzer module Chapter 3: Page Software configuration with TCT, continued Detector selection The following detectors are relevant to the Uras 14 analyzer module: Back to home page.
Shut off the gas supply sample gas and, if applicable, reference gas to the analyzer. There are 0 items available. Page Optical Alignment, continued Execution The “Geometric Addition” should be set to a minimum value using all calibration capabilities. Pressure sensor Accessories Multimeter Voltage Electrical value: Sample Cell Sample cell Location in module The sample cell is installed between the main frame with modulator and the infrared detector or calibration unit.
In this chapter you will find the following information: Page 20 Summary, continued Figure Analyzer module Uras 14 open 1 Calibration unit abb 2 Measurement cell 2 3 Main frame with aperture control 4 Beam path 2 aperture adjustment screw 5 Modulator with emitters not visible 6 Calibration unit 1 7 Infrared detector 1 8 Beam path 1 aperture adjustment screw 9 Measurement cell Solenoids Solenoids General To calibrate the analyzer module from one to three optional solenoids can be installed and then set up using the Gas urax menu item.
Remove the infrared detector wiring. Flow Detector Flow detector General An internal flow sensor can be used to measure throughput.
Contents, continued Page Chapter 5: