The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||22 July 2011|
|PDF File Size:||3.96 Mb|
|ePub File Size:||19.25 Mb|
|Price:||Free* [*Free Regsitration Required]|
The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of the CoaguChek XS monitor for both capillary and venous whole blood samples is very good. The letters mem The CoaguChek XS Meter automatically stores up to test indicate coagucnek you are viewing a results and their dates and times in its memory.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
Advanced Features Advanced Features 3. Allow to Dry With the cover off, allow the The CoaguChek XS Meter includes advanced features that you can test strip guide to dry for set up as necessary.
If you see the another method.
Get Ready Take a test strip out of the then press M container. For the capillary testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor.
Open Battery Compartment With the meter turned off, turn it over. To go to of each letter, number, and symbol appear.
Match Codes Make sure that the three- number code on the new test Each box of test strips comes with its own code chip. For more information, see re-insert it.
Page of Go. Erasing Stored Test Results 2.
Record Result Record the result according to test strip. A rotation scheme was used to ensure that the same instrument was not always dosed first. If you see an error message, first try to warm to measure correctly. Cleaning the Meter It is important to keep the coagufhek clean.
Preparing For A Test 1. Page 26 Repeat the test. Page 99 The Coefficients of Variation presented for precision of whole blood duplicates indicate that the pre- cision of voaguchek CoaguChek XS monitor for both capillary and venous whole blood samples is very good. Repeat the test using a new test Error: Match the Code Confirm that the number Performing a Test displayed matches the number on the test strip container, 1.
Indicates the results are displayed in Seconds. Test Strip Interference Solution: Gather Items Gather the following coaguchsk Using Venous Blood 2.
For the venous testing, blood was applied from a syringe; There is a potential risk of infection. Got it, continue to print. Table Of Contents User Manual Clean the meter whenever Cleaning the Test Strip Guide it looks dirty or, if you prefer a regular schedule, clean the meter each time you open a new box of test strips.
Code Chip Code Chip 2. Performing A Test 3. The CoaguChek XS PT test strip may be used for patients manhal a combination therapy of oral anticoagulants plus heparin injections. Dose the Test Strip Discard the first 4 drops of blood. Operating Conditions Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test.
Don’t show me this message again. By pressing ‘print’ button you will print only current page.
Roche CoaguChek XS User Manual
All other trademarks are the property of their respective owners. Fill the Capillary Tube When the flashing test strip 4. Don’t have an account? Blood 1 hanging drop of blood may be collected directly using a butterfly or a needle and syringe. The correlation was 0. Page 91 For the capillary testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor.
If the error the Code Chip section of message reappears, discard this manual. Tables 1 and 2; a visual representation is provided in Graphs 1 and 2. Keep the tube level, and allow it to fill by capillary action. If the memory is result in memory. Page 24 For more information, see re-insert it. To coaguchsk Erase mode without To erase maunal stored results: Integrated Quality Controls Integrated Quality Controls Tips for a Good Fingerstick The CoaguChek XS System has quality control functions integrated For fingerstick blood testing, increasing the blood flow in the finger into the meter and test strips, so you never have to run quality will help you get a good drop of blood.