Fluke A Multimeter DIGITAL | Business & Industrial, Test, Measurement & Inspection, Test Meters & Detectors | eBay!. The listing you’re looking for has ended. Fluke A Multimeter. Condition: For parts or not working. Ended: Jun 08, , AM PDT. Price: US $ Corporation (Fluke) has reproduced this copy of a manual which is no All shipments of JOHN FLUKE MFG. CO. . A A rechargeable Ni Cad battery pack.
|Published (Last):||5 August 2013|
|PDF File Size:||4.78 Mb|
|ePub File Size:||7.1 Mb|
|Price:||Free* [*Free Regsitration Required]|
Full text of “FLUKE a”
When the probe tip is applied to a solid surface it draws or sinks heat from the surface. Change the voltage applied to fpuke V-fl input terminal to 1. Similar sponsored items Feedback on our suggestions – Similar sponsored items. Exercise the A in the following manner and observe the oscilloscope for the appropriate waveform. Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to watch list.
Fluke A Multimeter DIGITAL | eBay
This includes items that tluke defective in ways that render them difficult to use, items that require service or repair, or items missing essential components. Each listed part is shown in an accompanying illustration. Have one to sell? Autorange is not working. As expected, it was mainly a cleaning job for the switches and few resoldering around the main switch panel to get this back to working. Fljke, VDC function, 2 ranges, 2.
The following instructions should be adhered to in order to obtain the best results from the 80T probe. Set the Fluie, front panel up, on a flat surface and gently lift the bottom edge of the front panel up to clear the switch buttons. Physical design of the Model 80F-5 allows direct mating to the input terminals of the Fluke voltmeters.
Select the dc mA function and 2 range. Change the voltage applied to the V-fl input to 1. Select the 8040 range. Note the resistance value indicated on the A display. Since I am slightly sick and do accumulate all sort of test instrument garbage, I paid for this and got it home.
Probable cause, U22 or DS8 defective.
Apply volts dc to the Model 80F high voltage probe. Add to watch list Add to wish list. Interest will be charged to your account from the purchase date if the balance is not paid in full within 6 months. A mV and 2V ranges: The A display should read between 1.
This list presumes an availability of common electronic parts at the maintenance site. By continuing to use this website, you agree to their use. Special financing available Select PayPal Credit at checkout to have the option to pay over time. The location of each calibration adjustment is illustrated in Figure NOTE If the display does not stop changing center R88 then return to step 13 and repeat through step This procedure may be used as an instrument operational flke when the A is being used in locations away from normal calibration equipment.
When the full scale capability of the selected range for any function is exceeded, the display will blink.
The reference voltage is selected, by the Timing and Control circuit, to be opposite in polarity from the output of the Buffer or RMS Converter.
Report item – opens in a new window or tab. Insure that the A2 and A3 boards fit into the plastic board edge retainers on each side. No additional import charges at delivery!
Resistance Measurements The Model 80F-5 division ratio should be checked every month or every year depending on the accuracy requirement. When reassembling the A, protect the flexible lands by first folding the switch section of the A1 pcb then the display section up towards the vertical A2 and A3 pcbs.
It is recommended that the A circuit board remain folded up and attached to the front panel. The steps of this procedure must be performed in the order presented.
Shipping cost cannot be calculated.
John Fluke Model 8040A Digital Multimeter
Leave a Reply Cancel reply Enter your comment here Instrument Model and Revision Letter. Electrical components are listed by reference designation and mechanical components are listed by item number. Figure defines the rf signal limits that can be tolerated without degrading measurement accuracy. Where Vi is the value of the ac voltage at any given instant. Mouse over to Zoom – Click to enlarge. NOTE If the amplitude of the first ms of either oscilloscope display significantly deviates from waveform A3 or B3, U2 is the probable cause of the malfunction.
Fluke 8040A Multimeter
Therefore, if the measured surface has a low mass e. The four non-rechargeable alkaline batteries are to be mounted in the battery clips position as indicated on A case then the battery cover secured in place with the two screws provided. Clean dust from the circuit board with low pressure dry air. The output is positive polarity dc which is calibrated to be equivalent to the rms value of a sine wave.
Install the Model 80F-5 front cover and strain relief.