Read online or download PDF • Page 44 / 67 • KROHNE IFC Converter EN User Manual • KROHNE Equipment. Micro Precision provides expert calibration services for KROHNE instruments including KROHNE IFC D -. Select from options for the IFC D which. KROHNE 08/ Subject to change without notice. Applicable to. Software-Versions. ○ IFC _. / D. Display version. No. and.
|Published (Last):||28 October 2016|
|PDF File Size:||8.36 Mb|
|ePub File Size:||11.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 10 3 Start-up 3. Page 15 Function 3. If installed and operated in accordance with these operating instructions, it will rarely present any problems. Page 8 Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical connection of outputs 2.
Page 2 Contents 1 1. Your manual failed to upload Vo r tex flowmeters.
V ariable area flowmeters. The operator is, howeverobligated to take note of these sections in the. The selected creepage distances and clearances have been sized in conformity with VDE and IEC for contamination category 2. Page 14 Function 1.
What’s missing? Tell us about it.
Passive mode External power source required for operation of the outputs. Supply circuits and output circuits are designed to meet the standards of icf classes III and II, respectively. Use separate cable entries see below for power supply, field current cables, signal lines, outputs and inputs. Or point us to the URL where the manual is located.
Page 11 Standard factory settings Function 1. Switches, counters, displa ys and recorders.
KROHNE Aquaflux K/D/6 with Signalkonverter IFC D Unused | eBay
Do not cross or loop cables inside the terminal compartment. These concise instructions do not include the following: Power the unit, and the flowmeter will immediately start process flow measurement.
Underwater and underground installation possible.
P Pulse output overranged. All operating data have been factory set in accordance with your specifications. Lev el measuring instruments.
KROHNE IFC Converter Short EN User Manual | 19 pages
Don’t see a manual you are looking for? Tell us about it.
Signal cable to be permanently laid. Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical connection of outputs 2.
Setting ranges The ranges are dependent on the meter size DN and the flow velocity v: Connect shields via stranded drain wires. The flowmeter housings protecting the electronic equipment from dust and moisture must always be kept closed.
Page 3 1 Electrical connection: Tell us what’s missing.